Manual zz. If you use a calibration kit other than a predefined one, you need to define it. If the connector type of the standard of the calibration kit you use has polarity the distinction between male and female , you need to change the standard class definition of the calibration kit depending on the standard you actually use. For more information, see Modifying Calibration Kit Definition. This calibration effectively eliminates the frequency response reflection tracking error, directivity error, and source match error from the test setup in a reflection test using that port. Press Cal Key.
|Published (Last):||17 April 2008|
|PDF File Size:||1.75 Mb|
|ePub File Size:||6.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Manual zz. If you use a calibration kit other than a predefined one, you need to define it. If the connector type of the standard of the calibration kit you use has polarity the distinction between male and female , you need to change the standard class definition of the calibration kit depending on the standard you actually use. For more information, see Modifying Calibration Kit Definition. This calibration effectively eliminates the frequency response reflection tracking error, directivity error, and source match error from the test setup in a reflection test using that port.
Press Cal Key. Click Cal Kit, then select the calibration kit. If the name label of the calibration kit has been changed, the label is displayed as the soft key. Click Calibrate. Select a test port and corresponding S parameter on which 1-port calibration will be performed. Click Open to start the calibration measurement. Click Short to start the calibration measurement.
Click Load to start the calibration measurement. Click Done to terminate the 1-port calibration process. Upon pressing this key, calibration coefficients will be calculated and saved. The error correction function will also be automatically enabled. This calibration effectively eliminates the directivity error, crosstalk, source match error, frequency response reflection tracking error, and frequency response transmission tracking error from the test setup in a transmission or reflection test using those ports.
This calibration makes it possible to perform measurements with the highest possible accuracy. A total of twelve error terms, six each in the forward direction and the reverse direction, are used in a calibration. Press Cal key. Click Select Ports, then select the test ports on which you will perform full 2-port calibration.
In the procedure below, the selected test ports are denoted as x and y. Click Reflection. Click Port x Open to start the calibration measurement x denotes the test port to which the standard is connected.
Click Port x Short to start the calibration measurement x denotes the test port to which the standard is connected. Click Port x Load to start the calibration measurement x denotes the test port to which the standard is connected. Repeat the above procedure for port y.
Click Return. Click Transmission. Click Port x-y Thru to start the calibration measurement x and y denote the test ports between which the THRU connection is being made.
If an isolation calibration must be performed using a LOAD standard, follow the procedure below. Click Isolation Optional. Click Port x-y Isol to start the calibration measurement x and y denote the port numbers to which the LOAD standard is connected.
Click Done to terminate the full 2-port calibration process. Safety Safety Summary The following general safety precautions must be observed during all phases of operation, service, and repair of this instrument.
Such noncompliance would also violate safety standards of design, manufacture, and intended use of the instrument. Agilent Technologies assumes no liability for the customer's failure to comply with these precautions.
Operation of any electrical instrument in such an environment clearly constitutes a safety hazard. Keep Away from Live Circuits - Operators must not remove instrument covers. Component replacement and internal adjustments must be made by qualified maintenance personnel.
Do not replace components with the power cable connected. Under certain conditions, dangerous voltage levels may remain even after the power cable has been disconnected. To avoid injuries, always disconnect the power and discharge circuits before touching them. DO NOT Service or Adjust the Instrument Alone - Do not attempt internal service or adjustment unless another person, capable of rendering first aid and resuscitation, is present.
Dangerous Procedure Warnings - Warnings, such as the example below, precede potentially dangerous procedures throughout this manual. Instructions contained in the warnings must be followed. Dangerous voltage levels, capable of causing death, are present in this instrument. Use extreme caution when handling, testing, and adjusting this instrument. Safety Symbols Instruction Manual symbol: the product is marked with this symbol when it is necessary for the user to refer to the instrument manual.
Alternating current. Direct current. On Supply. Off Supply. In-position of push-button switch. Out-position of push-button switch. A chassis terminal; a connection to the instrument's chassis, which includes all exposed metal structure.
Related manuals Network Analysis Back to Basics. Keysight Vector Network Analyzer. EA - Keysight. Network Analysis Back to Basics. TR Operating Manual. EA User's Guide. Planar R54 Operating Manual. R Operating Manual. D Softkey Functions. Download PDF advertisement. Its a community-based project which helps to repair anything.
E5071C User Instructions
Using this Help To understand the conventions and navigation system used in this help system, refer to Using this Help. To know about the new features of the EC, refer to What's New. Help Revision Date A. Using this Help This help provides the user and programming documentation in an searchable electronic format for the EC Network Analyzer.
Help File - User Manual for E5071C ENA Network Analyzer (Operation and Programming)
Aglient EC This document contains proprietary information that is protected by copyright. All rights are reserved. No part of this document may be photocopied, reproduced, or translated to another language without the prior written consent of the Agilent Technologies. The printing date changes when a new edition is printed. Minor corrections and updates that are incorporated at reprint do not cause the date to change.
Keysight e5071c manual
See search results instead:. Sprechen Sie mit einem Experten. View Product Details. Alle detailierte Suchkriterien entfernen.
Agilent Technologies E5071C Manual